Payment & Shipping Terms:
|Lifting Force (at Any Power Level):||≥70N||Sensitivity (at Any Battery Level):||A1-15/100|
|Integrated Electromagnetic Probe Volume::||165*148*45mm||Center Distance Between Magnetic Poles:||10～170mm|
permanent magnet integrated ndt equipment,
ndt magnetic flaw detector,
non destructive testing magnetic flaw detector
Permanent Magnet Integrated Yoke Flaw Detector HCDX-DA172
1. This product is a wireless integrated machine composed of a miniature yoke and an AC magnetization power supply, without any external wiring, integrating magnetic detection, power supply, lighting and other functions in one, compact structure, small size, light weight and easy to use. The defects such as disconnection and unsealed plug are solved.
2. This product can generate a stable AC magnetic field without an external power supply, ensuring detection sensitivity, and magnetic particle inspection is simple, convenient, efficient and safe.
3. This product integrates an inverted electromagnetic field intelligent constant magnetic chip, intelligently monitors and dynamically adjusts the electromagnetic intensity, so that the electromagnetic output is kept in the best state, and the lifting force and sensitivity are not affected by the battery power. Ensure the quality of inspection.
4. The instrument imitates ergonomic design and feels comfortable.
5. The instrument will automatically shut down if it is not working for a long time when it is turned on, which saves power and is safe.
6. The lithium battery can be replaced at will, which solves the worry of insufficient power supply.
7. The yoke probe of this product integrates a high-brightness LED white light source, which can be used for illumination and inspection image observation.
8. Intelligent multi-purpose switch, one-key operation of the whole machine, not only convenient to use, but also reduced failure points.
9. Intelligent no-load automatic identification and control function, avoiding the over-current and overheating of the instrument when it is no-load.
10. Intelligent soft start and soft stop functions, buffer the electrical shocks generated by the instrument during frequent starts and stops, and prevent electrical shocks from damaging components.
11. Automatic monitoring and protector for battery operation, in case of abnormal conditions, cut off the circuit in time to prevent accidents.
12. The power indicator indicates the power level in real time.
13. The special probe foot adjuster makes it convenient, easy and fast to adjust the probe foot.
14. Equipped with a dedicated work package, work easier and safer.
Lifting Force (at any power level): ≥70N
Sensitivity (at any battery level): A1-15/100
Integrated electromagnetic probe volume: 165*148*45mm
Weight of integrated electromagnetic probe: 1.1kg
Center distance between magnetic poles: 10 ~ 170mm
Lithium battery module volume 52*45*86mm
Lithium battery module weight: 200g
White light intensity: p 2000 lux
Work duty cycle: 100%
Full power working time: 2 days
Battery configuration method: replaceable
Charging Time: 4 ~ 6 hours
Dedicated charger: 16.8V/1A/100 to 240VAC
Integrated yoke (without battery) 1
2 dedicated batteries
2 dedicated chargers
Foot adjustment tool 1
Oblique feet 1 pair
Dedicated work bag 1
Instruction manual 2 copies
Certificate of conformity 1 copy
Instrument case 1
Lifting force test block
Magnetic particle sensitivity test piece
How to install the battery:
1. Install the battery;
2. Long press the "start switch" for more than 0.5 seconds, and the white light will stay on for 3 seconds to indicate successful power-on;
3. After successful power-on, if there is no work, the white light will flash every 3 seconds;
4. After power on, the magnetic detector will automatically power off if there is no operation for more than 2 minutes.
Power off method:
There are two ways to power off:
1. Unplug the battery directly;
2. If the magnetic detector is not operated for more than 2 minutes, the power will be cut off automatically.
Flaw detection operation method:
1. Fully charge the battery and install a fully charged battery;
2. Use the foot adjuster to adjust the proper foot spacing;
3. Long press the "start switch" for more than 0.5 seconds to turn on the power;
4. If it is a model with UV lamp, switch the black and white light according to the needs of flaw detection;
5. Press the "start switch" to perform AC magnetic particle inspection, and the black and white light will automatically light up;
6. Release the "start switch" to stop the magnetic particle inspection, and the black and white light will go out;
7. After the flaw detection is over, just make sure that the "start switch" has been released. At this time, when the magnetic detector is not operated for more than 2 minutes, the power will be automatically cut off.
Remaining battery check:
1. The remaining power can be checked when the battery is alone or installed on the electromagnetic probe;
2. The battery indicator is on the top of the battery. A single vibration will trigger the internal switch and the battery indicator will light up;
3. Determine the remaining battery power from the table below according to the color of the battery indicator;
4. The different display modes of the power indicator indicate different remaining power, as shown in the following table:
|mode||remaining battery||Battery indicator working mode|
|Battery level display during charging||C<30%||Red light, flashing|
|30%≤C<70%||Orange light, flashing|
|70%≤C<100%||Green light, flashing|
|C=100%||Green light, always on|
|Battery display when not charging||1%<C≤20%||Red light, flashing|
|20%<C≤50%||Red light, always on|
|50%<C≤75%||Orange light, always on|
|75%<C≤100%||Green light,always on|
1. Please do not work with no load for a long time. When the remaining power is insufficient, please remove the battery and charge it with the supporting special charger. When the charger indicator turns from red to green, the charging ends.
2. The instrument has been discharged when it is shipped, so please charge it before using it for the first time. If you need to travel long distances, please run out of electricity, which is safer.
3. The instrument is extremely sensitive to the charging voltage and current, and the original charger of the machine must be used for charging. If you use other chargers to charge the battery of this machine, there will be potential safety hazards. The company will not be responsible for all consequences caused by this.
4. Both the illuminating lamp and the ultraviolet lamp are strong light sources, so direct eyes are forbidden;
5. Ultraviolet light will harm the human body to some extent, so avoid direct exposure to the human body;
6. If this product is not in use temporarily, please charge the power supply with 50%-60% power, and then store it in an environment where the temperature is 20℃±5℃, the relative humidity is 45%-85% and children are not easy to reach.
HUATEC has always been adhering to the first concept of customer-oriented, the products we sell will be carefully checked before shipping and then delivered to your hands.If you have any product requirements, you can contact me.Looking forward to your order.WeChat:19520504939
Contact Person: Mr. Davis Chen