metal hardness tester,
steel hardness tester
Shore Durometer,Shore D hardness tester, Shore D durometer HS-D
HS-D Durometer is an essential characteristic of various hard rubber, colophony, glass, print board, fiber and so on. It is feature portability, ease of operation, with facility, high resolution and it is available in GB/T 531 -1999, and ISO 7619, ASTM D2240.
Capacity: 0~100 HD
Available test range: 20~90 HD
Tip stroke: 0~2.5 mm
Tip dimension: SR 0.1mm
Net weight: 0.2kg
Shore durometer is used to test hardness of sulfureted rubber and plastics, such instrument has A-type/D-type/C-type.
A-type and D-type apply to test the low and medium level hardness and high level hardness of materials.
C-type applies to test the hardness of shoe-making porous materials made by vesicants within plastics when the compression rate is 50% and stress is beyond 0.049MPa.
Put the sample on a solid plane, hold the durometer to make the indenter at least 12mm away from the sample edge, and stably put the legs of durometer on the sample. Thus the indenter straightly goes into the sample till the legs completely get to the sample. Check the reading in 1s.Repeat such test 5 times at different place which are at least 6mm away from the test point and find the average reading (the distance between porous materials is at least 15mm).
Contact Person: Mr. Davis Chen