|Test Range:||0.0-300.0（Ｇs）||Test Accuracy:||＜1.2（Ｇs）|
75 Degree Gloss Meter,
Gloss Meter 75 Degree,
Non Destructive Testing Gloss Meter
High Precision 75 Degree Paper Gloss Meter HGM-BZ75 For Paper Testing
International standard ISO-8254, TAPPI - T480, ASTM-C346, ASTM-D2457, DIN67530 and China standard GB8941.3
The technical parameters follows China metrology standard JJG696-2002.
BZ75 ° intelligent glossiness meter is follow ISO - 8254, TAPPI - T480, GB8941.3 standard manufacture high-precision, miniaturization, intelligent paper with glossiness plan, suitable for paper, etc, the paper is printed paper surface gloss surface lens to value measurement.
|The technical parameters up to JJG 696 (First-level standards).|
1. Small and light, real easy to be carried and used.
2. 4.8V Rechargeable battery can be used for almost 60 hours and 10000 reading.
3. High stability
4. Long-term stable light source needs no replacing forever.
5. Quartz crystal standard board, no excursion.
6. Single-chip microcomputer control circuit, the new keystrokes, light and portable, a single calibration, intelligent calibration record values
Use of equipment
Press the power switch in the measurement mode, apparatus on the sample, the samples showed that the numerical point of view at this time of the gloss value (Gs). When the "angle/ shift" button, switch different angle measurement, the corresponding point of display lights. Whether measured by that point, when measured value greater than 300.0 Gs, shows that "OVER" word.
When the electricity shortage Show "bAtt" words, at this time should be promptly charged with exclusive power rechargeable battery, charging devices plug into the mains socket on the side of charge, charging time in about four hours. Charging standards "⊕", charging the panel led by lit.
When standard equipment on board showed that the standard plate with the corresponding numerical point of difference is greater than the nominal value of 1.2 gloss units at the following methods should be calibrated:
And measuring window on the black velvet board, press the "load switch / zero setting" button, this shows that numerical 0.0.
Black glass plate instead of the standard black velvet boards, measuring instruments on the window, rather than pressing the "calibration / test" button to enter the set, and then display the "000" and the first "0" flashing. Rather than pressing the "angle / shift" key-flashing Move right, rather than pressing the "load switch / zero setting" key "flicker" of numerical change from 0-9-0 through the press "angle / shift" and "load switch / zero setting"key to making shows that numerical standards and black glass plate corresponding nominal value of the same, at this time rather than pressing the" calibration / test "button to complete the setup. And white ceramic standard equipment on board, show that numerical standards-and-white ceramic plate angle of the corresponding nominal value of the error should not exceed ± 1.2 gloss units, it should be considered standard plate dirt, and other issues. Which point of error of work, occupation which should be in work mode in accordance with the perspective of the above process of calibration. After the completion of the steps of the samples can be measured.
The equipment built four rechargeable batteries,
A black board is used for the highlight calibration
A white board is used for the low light calibration
|English instruction manual||1pc|
HUATEC Gloss meter Reference angle for different application
|angle||Point of reference for (foreign) (China)|
|high gloss paint, ink, plastic.|
|high gloss stencil, cast coated paper.|
|ISO-2767 GB8941-2||surface of aluminum and aluminum oxide.|
|pottery, ceramic, plastic, plastic film, Stone|
|45°||JIS-E8741||pottery, ceramic, plastic, plastic film, Stone|
Mid-gloss paint, ink, plastics, ceramics, stone.
HUATEC has always been adhering to the first concept of customer-oriented, the products we sell will be carefully checked before shipping and then delivered to your hands.If you have any product requirements, you can contact me.Looking forward to your order.WeChat:19520504939
Contact Person: Mr. Davis Chen